A long time ago, around 12-13 years, I build one of the classical designs of a laboratory power supply and quickly started thinking about having a digitally controlled one. I got a prototype working in 2012, but made no progress after that, seemed my simple 7805 & LM317 was covering me for most part, so a total lack of motivation presented itself to finish the project. However, suddenly realizing I have wanted this thing build for more than 10 years got the gears working, it has to be finished!
I have improved the previous design by going up for higher speed current sensor (INA225) and faster amplifiers, see Electrobob 2 below. Dropped the current mirrors, since there was no need for more than 30V output. Still, I was not satisfied with the speed. In search for the perfect recipe, I decided to analize the various concepts and make a short summary of performance and lessons learned.
I have drawn simplified schematics to explain how each supply works. Some changes were made to the original schematics where needed to reduce the requirement of output capacitance to just 10uF. The results are simulation based, with the exception of Electrobob 1 and Electrobob 2 which were built and measured.
1. My previous design – Electrobob 1. It uses a single supply, no additional negative/floating rails. The positive rail current sensor allows for multiple channels to share a ground. It is designed to scale in voltage and current easily by using current mirrors to control the main transistor. Short circuit response is slowish because the used current amplifier(ZXCT1010) is rather slow. Therefore:
2. An improved design – Electrobob 2 – keeping the same features as before: went for a faster INA225 current amplifier. The current mirrors are missing, because the faster LM7332 can operate to over 30V. With these design changes and proper compensation, it is 10 times faster to respond to a short circuit compared to Electrobob 1.
4. The most CLASSICAL design to make a great power supply, however it requires a symmetrical (Vpos, Vneg) isolated power supply apart from the main Vin. The ground reference is the positive output, and Vref and Iref need to be generated with respect to this one. It is found in a lot of designs and sold by a lot of companies. Provided Vpos/Vneg are clean (easy to do because of low power) it has very good rejection of Vin. Due to the extra supplies, it can have a low drop across the series transistor and can scale easily in voltage.
5. The simplest design I can think of – SE simple – normally found in many regulators: uses current sensing on the negative rail, so this voltage is fed back in the voltage reference to compensate (R6-7). Vref and Iref are ground references, therefore it avoids extra current through the Shunt as with design 3. Speed depends largely on the speed of the op-amps used. The topology does not need any additional negative supply, provided that the op-amps output can reach ground.
7. The classical LM317 regulator, along with a current sensor and comparator, to get it working in constant current mode as well. It requires a negative supply, as ADJ needs to be lowered to -1.25V in order to achieve 0V output in short circuit.
The table below measures performance with some simple tests. The simulation included secondary effects such as wire resistance and inductance, capacitor ESR, contact resistance, feedback and source resistance.
The load test is measured with the output set to 5V, a 1A load is connected and disconnected. The Drop voltage and Drop time are measured, when the load is connected. The opposite Peak voltage and Peak time are measured when the load is disconnected. Ideally the voltage should not drop or peak when the load changes, however in practice the peak and drop should be as small and short as possible. Output resistance was measured, but it is not relevant for comparison, it is within the mΩ range for all designs except LM317.
The short circuit test shorts the output and measures the peak current and the time until it drops close to the expected current limit. The current should never peak above the preset limit, however this does not happen because of 2 reasons: the output capacitor will provide some current and the current control loop takes some time to kick in. Depending on the configuration and the type of output transistor, the peak current at short can be rather high.
The PSRR is measured by aplying an AC voltage an the input of the supply and measuring the output, over frequency, as in the example below
The table shows the comparison of the discussed models and points out their strengths and weaknesses.The peak current indicates the peak through the pass transistor, while the load current is higher due to the output filter capacitor.
- When connecting/disconnecting 1A load.
- Is oscillating a bit, dampened in 10-15us
- Goes to zero and then comes back up
- Using TIP122, caused by high drive capability of LM7332. Can be reduced with extra R+NPN, IRLZ24 – about 60A
- Using IRLZ24 MOS, with TIP122, the drop/peak are higher, ~1.4V due to different gains at different currents (BE resistors), but peak currents are lower. Can be dangerous for the output circuit!
- compared to main supply input
- i don’t know why
- Requires separate secondary for +/-5..12V. However, cheap +/-5V 1W isolated DC/DC converters could solve this, like DET01L-05
- Using LM358, 5µs with TL3472
- Using LM358, 20µs with TL3472
Maybe it is time to revisit the idea of positive rail current sensor and go for a simpler, faster and cheaper approach, even though this might mean breaking the ground with the shunt.
A bit of history
Electrobob 1 was previously tested as V1, and with some updates it turned into V2 which never made it to the tests. Then I did some radical changes and it turned into V3, which was a failure, both as stability but also PCB design. Here it is below trying to split it in 2 boards.
Next came version V4, which is actually Electrobob 2, as presented above. I made a lots of progress with this one:. It fits in a 5×5 PCB without holes, which turned into a 5x7PCB with holes. I made a prototype with a screen and box, to be used with an external power adapter. It has some cool features like temperature protection, OVP and lots of software features.